"Saddal Surgical" is an established name in the Manufacturing sector having built its reputation on housing the finest quality products such as Surgical, Dental, Beauty & Veterinary Instruments.
Deposits or Organic Substance.
Rinse instrument thoroughly and immediately after use in distilled or de-mineralized water.
COLORED HALO STAINS AND THE SHARP CONTOURS ARE NOT PRESENT
In flushing water or steam there are iron, copper, manganese etc: are present.
Use distilled or de-mineralized water.
Check Composition of Cleaning Solution
Old cleaning solutions/ Incorrectly cleaned instruments
Change baths after each use/ Rinse instruments immediately after use in distilled or de-mineralized water.
Detergent with low pH content ( 6 or less )
Use Neutral Detergent
Metallic Coating or Platting
Metals of different types have been sterilized together such as: for instance stainless steel with Silver / Chrome / etc, etc.
Sterilize only the similar metal instruments together.
Excess Heat (Chromatic Oxide)
The autoclave is not operating properly or has been set to very high temperature (This can cause metal to loose some of its properties such as strength and hardness etc, etc.)
Correct the autoclave temperature
ADDRESS: 100 Tenth Avenue, New York City, NY 1001 FOR BOOKING: (044) 359 0173
MONDAY - FRIDAY 11:30AM - 2:00PM SATURDAY - SUNDAY 4:30PM - 10:00PM MONDAY - FRIDAY 5:30PM - 11:00PM
There are design companies and then there are user experience design professional.